Kazragami The reading on the bridge is approximately 9. The modulated signal is transformer coupled to a demodulator driven at the same 3 kHz rate as the modulator. The Recorder Output Option Checks require the use of a dc differential voltmeter. DC Voltage Range Calibration To calibrate the Kelvin-Varley Divider, apply ac power to the instrument and proceed as follows: Transistor Q8 is driven in Synchronism with the chopper Ql and Is operated in the inverse mode to secure minimum saturation voltage. Tektronix D1 Curve Tracer.
|Published (Last):||4 November 2017|
|PDF File Size:||4.49 Mb|
|ePub File Size:||5.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
Electrical Specifications Differential Voltmeter Conventional Voltmeter TVM General Specifications Mechanical Specifications. Operating Features Preliminary Operating Procedures Operational Test Ground Loop Currents Common Mode Voltages Use of Shorting Link TVM Mode Operation Optional Recorder Output Operation. Measurement of Extremely High Resistances.
Operation as a Null Detector Measurement of DBM Block Diagram Analysis Conventional TVM Mode Differential Null Mode Optional Recorder Output Circuit Analysis The Reference Supply Kelvin-Varley Divider AC Converter Null Detector DC Input Divider Service Information Disassembly Instructions The Model A Disassembly Instructions.
General Maintenance Periodic Cleaning Fuse Replacement. Battery Replacement Precalibration and Troubleshooting. Kelvin-Varley Divider Checks Recorder Output Option Checks. AC Converter Checks. Calibration Procedures Meter Mechanical Zero Adjust. Zener Reference Supply Calibration Null Detector Calibration Kelvin-Varley Divider Calibration.
DC Voltage Range Calibration AC Converter Calibration Column Description How to Obtain Parts List of Abbreviations Serial Number Effectivity
Fluke 893A Instruction Manual
RMS Differential Voltmeter 931B
Produkt: FLUKE 893A FLU-Q893A
Fluke 893A Manuals